The modern gynecological operating table uses hydraulic system, which has the characteristics of compact structure, small size, light weight, low noise, stable control, easy installation and overall layout. The hydraulic system includes fuel tank, check valve, solenoid valve, relief valve, capsule accumulator, pressure gauge, throttle valve, and cylinder. The capsule accumulator can be used as a power source to store pressurized oil and release energy when needed. Automatic compensation of pressure is achieved according to the pressure gauge. The drive motors are driven independently by their respective drivers. The side of the operating bed is equipped with hand controller, and various motion controls can be performed by the hand controller.
The actions of the gynecological operating table are controlled by microprocessor. The microprocessor is connected to the hydraulic and motor execution sections via control drive section, and the hydraulic and motor execution sections are used to perform various movements of the operating bed including: forward and backward movement, longitudinal movement, height adjustment, etc. The use of microprocessor design can simplify the circuit design, greatly reduce the line fault point and ensure the reliability of the action.